Nano vs Alibaba Hearing Aids Comparison. Nano Listening to Support vs Alibaba Amplifier Comparability | Excessive Tech Listening to Support or Low-cost Amplifier? Dr. Cliff Olson, Audiologist and Founding father of Utilized Listening to Options in Anthem Arizona, compares the Nano Superior Listening to Expertise Listening to Aids with a Low-cost $20 Alibaba Listening to Amplifier.
Nano vs Alibaba Hearing Aids Comparison
Web site: https://www.DrCliffAuD.com
Clinic Web site: https://www.AppliedHearingAZ.com
*The data introduced on this video is my very own opinion. My content material is meant to supply info that’s of worth to the general public on problems with public concern. I didn’t obtain any compensation for this video.
When you’ve ever completed a seek for on-line listening to aids after which gone on Fb, you’ve got possible been bombarded by adverts from an organization known as Nano Listening to Applied sciences.
In accordance with the Nano web site, they manufacture Superior Listening to Aids that “Auto-Regulate” to you listening to loss with out the necessity of a listening to check, they work for “98.7%” of people with listening to loss, and have over 3,000 opinions and not using a single 1 or 2 star evaluate, and all of this for under $400 for a pair of units.
Upon taking a look at these Nano listening to aids, they look like similar to the $18-20 amplifiers offered on Alibaba.com. Alibaba connects producers of low-cost merchandise from locations like China and India to distributors who sells these merchandise to shoppers.
An unboxing of the Nano Listening to Aids and Alibaba Listening to Aids revealed units that have been 99% an identical apart from the NANO branding and the Chinese language branding. Nonetheless, seems could be deceiving so we determined to objectively check the efficiency of the units to see if there are any variations.
The primary check is the Diagnostic check inside the Otometrics Listening to Instrument Check Field. This enables us to check the listening to units in a really managed surroundings that eliminates nearly each variable of testing. Outcomes of this check confirmed that each units carried out nearly an identical to one another. Not solely did they carry out the identical, however even the quantity management functioned the identical. The quantity on each the NANO and the Alibaba system solely had 2 settings, Most quantity & Minimal quantity. There isn’t a in-between quantity settings with these units.
Subsequent, we examined these units utilizing Actual Ear Measurement utilizing the Otometrics Aurical Probe Microphone Measurement (PMM) tools. This check will allow us to see how these units carry out inside a consumer’s ear. It should additionally inform us if a prescription for a Regular to Average Sensorineural Listening to Loss could be met with these units. Upon testing the amplification, the units carried out nearly an identical. This was not stunning contemplating their related efficiency within the Listening to Instrument Check Field. They have been additionally not capable of meet the consumer’s listening to loss prescription.
Not solely do these units carry out identically, they appear an identical, really feel an identical, and have equally unhealthy sound high quality for my part.
I sometimes attempt to stay as unbiased as attainable, and do not care when you purchase listening to aids from me, one other supplier, Costco, or on-line, so long as you’re making an knowledgeable determination. That being stated, I might not, in good conscience, suggest that anybody even strive a NANO or Alibaba listening to amplifier.